[ Model：LEVEL TESTER ][ Stock No：500 ]
- Very convenient for Level Sensitivity, Levelness inspection and Levelness examination.
- The core interval for this equipment is 500mm. Normal Level Sensitivity id displayed in 1m. On this equipment the dial gauge reading value multiplied × 2 will be the level sensitivity.
NOTE: Level and Dial Guage are not included. They have to be provided separately.
|Code No.||Nominal||floor surface Measurement
|Upper surface Measurement
|Core interval Distance
|What is Level Sensitivity||The inclination required to change the bubble chamber’s bubble graduation by 1. This means the height (Unit : mm) or Angle (Unit : seconds) in relation to base 1m.Height relations in relation to angle and base is as follows
Angle 1 seconds = 4.85μm per 1m
≈5μm per 1m
|Types of Levels||Division||Sensitivity (mm/m)|
|Type 1||0.02 (≈4″)|
|Type 2||0.05 (≈10″)|
|Type 3||0.1 (≈20″)|